In the ozone aging test, the evaluation of the test results mainly adopts three methods; one is to record the time when the sample begins to crack during the test. The second is to classify the crack of the sample, and the third is to test after the test. The samples are taken for photography and then evaluated intuitively. The latter two methods can only give a qualitative concept and take time (such as taking photos). After referring to the relevant technical information, we tried to use the method of measuring the modulus (different elongation) of the ozone aging tablets to evaluate the test results. This method is based on the fact that when the test piece is under attack by ozone in the stretched state, the rubber molecular chain is broken due to the reaction of ozone and the double bond of the rubber molecule until a characteristic crack perpendicular to the stretching direction is generated. During this aging process, the stress that the rubber limb specimen can withstand under constant elongation is getting smaller and smaller. Therefore, it was decided to develop a small tensile machine suitable for measuring the stress of the sample before and after the ozone aging test.
1. The electrical working principle of the small tensile testing machine (1) The basic principle of the electrical principle block diagram of the xK-1 1 automatic small tensile testing machine. When working, first adjust the display to zero value, and preset the tensile test piece to extend a, and start the start signal. The voltage signal output by the sensor first enters the filter to filter out all peaks and AC interference, then enters the amplifier Amplified, the digital signal output by it is displayed as a force value. Due to the start, the motor rotates the encoder disk to work, and the encoder disk has 50 through holes to form a binary cyclic code. Photoelectric coding is a kind of high-precision shaft rotation angle converted into coded electrical signals, that is, converted into corresponding digital quantity (length quantity) by optical conversion method. The control system controls the action of the relay according to the preset pulse value (pulse conversion length) to make the motor rotate or stop working.
(2) Technical performance:
Load: O ï½ž 3ON; Frequency: 50Hz; Precision: 2 Ã— 10-2N; Resolution: 1/200 (2 Ã— 10 N per word); Power supply voltage: 220V Â± 10; Test piece length: 40toni (variable) Maximum display value: 1999; sample tensile length grades: 3O, 5O, 7O, 1O0; display mode is light-emitting diode display; primary element uses tension and pressure sensor, which can work continuously.
2. Mechanical principle of small tensile testing machine This machine uses a reversible motor to drive the screw, and the bar to drive the sliding sleeve and the cross bridge, so that the lower splint can be moved up and down. When the motor rotates the lower splint in the forward direction, the lower splint moves downwards, and the sample is stretched, which is obtained by the pull and pressure sensor above.
Butter smearing is a troublesome thing. As we all know, butter is stored in cold storage. It is cold and hard when it is just taken out. At this time, we need a Butter Knife to assist.
Our butter knives are bright and colorful, exquisite in workmanship, durable and of various styles, including classic and multifunctional ones. The classic butter knife is delicate and small, with thick and smooth edges. It can cut hard butter neatly and evenly quickly, so it tastes good. The multifunctional butter knife can cut butter into pieces, or scrape it into thick silk and thin silk to meet your various requirements.
Butter Knife,Bread Butter Knife,Stainless Butter Knife,Stainless Steel Butter Knife
Jieyang Huiyi Hardware Products Co., Ltd. , https://www.chinagdhuiyi.com